The HB Pulse Oximeter provides a simple universal tool for patient monitoring – from spot checks to continuous monitoring. All monitoring data can be. The Edan HB Pulse Oximeter includes a choice of Reusable Edan SpO2 Sensor (recommended standard: Adult), four (4) AA Size Batteries (# ). Product Description. The HB employs EDAN SpO2 technology. Its pocket- size design makes it extremely suitable for spot-checks and EMS use. Compact & .
|Published (Last):||14 September 2008|
|PDF File Size:||1.30 Mb|
|ePub File Size:||4.78 Mb|
|Price:||Free* [*Free Regsitration Required]|
But the visual alarm is still active. HB Pulse Oximeter User Manual system for seconds the pause period can be set to 30, 60, 90 or s by the user, see section 3.
If service is necessary, contact qualified service personnel or your local EDAN representative. If you do not hear the POST pass tone, it indicates the alarm system oximefer not work well.
Pulse oximetry works by applying the sensor to a pulsating arteriolar vascular bed, such as a finger or a toe.
Orange indicates the device is being charged.
The icon displays in information area. Green indicates that the charging is complete. Edzn Used in this Manual This guide is designed to give key concepts on safety precautions. The safety and maintenance check can be conducted by personnel from this company. Physiological alarms Physiological alarms, also called patient status alarms, are triggered by a monitored parameter value that violates setup alarm limits or an abnormal patient condition.
An extension cable can g100b used between the oximeter and the SpO sensor. Civic Office, or the shop where you purchased the product.
The sensor contains a dual light source and a photonic detector. Only the battery of mAh capacity meets the UL requirements.
HB Pulse Oximeter User Manual A functional tester cannot be used to assess the accuracy of eean pulse oximeter probe or the pulse oximeter monitor. Page 3 The user shall understand that eda in this manual grants him, expressly or implicitly, any right or license to use any of the intellectual properties of EDAN.
Failure to cover the sensor site with opaque material in high ambient light conditions may result in inaccurate measurements. EDAN will not provide a substitute product for use when the defective product is being repaired.
Edan HB Pulse Oximeter | eBay
Oximetr may occur in the vicinity of equipment marked with the following symbol: Table of Contents Add to my manuals Add. Introduction and edsn rate of adult, pediatric or neonatal patients in hospitals, intra-hospital transport and hospital grade facilities.
The oximeter does not require calibration. Set the SpO alarm limits h10b follows: Field strengths from fixed RF transmitters, as determined by an electromagnetic The latest 10 minutes trend graph and table of SpO 2 and Pulse Rate can be reviewed in the oxiimeter.
Choose this item by pressing the Function button, a dialog box pops up as follows Install the battery compartment cover. EDAN may define as user serviceable. Do not use the It can perform spot and continuous measurement for a short time.
You can set the Sensitivity to Hi or Low via the menu. Our processing team may contact you with an additional shipping cost prior to processing your order for shipments outside of the continental US this includes Hawaii, Alaska, and Puerto Rico. HB Pulse Oximeter User Manual Wavelength The sensor contains LEDs that emit red light at a wavelength of approximately nm and infrared light at a wavelength of approximately nm.
If the measured field edn in the location in which the oximeter is used exceeds the applicable RF compliance level above, the oximeter should be observed to verify normal operation.
Disconnect the oximeter and sensor from the patient throughout magnetic resonance imaging MRI scanning. The sensor unconnected icon and associated alarm indicate the oxmieter has disconnected or wire fault. Display numeric and Waveform of SpO 2 simultaneously. Feel free to email or call us to get exact shipping costs prior to placing your order.
To avoid shock, the caregiver should not hold the oximeter while using a defibrillator on a patient. HB Pulse Oximeter User Manual than the recommend separation distance calculated from the equation applicable to the frequency of the transmitter.
In measurement state, if there is no measurement data and no operation for 10 minutes, the oximeter will oxometer off automatically. If abnormal performance is observed, additional measures may be oxiemter, such as reorienting or relocating the oximeter.
If the ID exists, the following confirmation dialog box will pop up. Comments to this Manuals Your Name. Edan HB Pulse Oximeter. Page 90 HB Pulse Oximeter User Manual interruptions, it UT for 5s is recommend that the product be powered from uninterruptible power supply or a battery. Adultfour 4 AA Size Batteries Alternatively, you can send an email to EDAN service department at: Return pluse the previous menu by pressing Return in the menu.
The recommended disinfectants include: The customer or the user of the oximeter should assure that it is used in eddan an environment.
Edan H100B Pulse Oximeter Features:
oxieter Page 92 HB Pulse Oximeter User Manual than the recommend separation distance calculated from the equation applicable to the frequency of the transmitter. Battery Charger Stand Kit. It is reminded that the product shall be used strictly complying with this manual.