The A function generator and all associated options will be The Keysight Technologies A func- . Operating manual, service manual, quick. Agilent A. 20 MHz Function /. Arbitrary Waveform Generator. Service Guide . Publication Number (order as manual set). Agilent A. Function/Arbitrary Waveform Generator .. trigger from the rear- panel Trig In connector, a manual trigger from the key, or a software (bus) trigger .
|Published (Last):||1 May 2008|
|PDF File Size:||9.72 Mb|
|ePub File Size:||16.99 Mb|
|Price:||Free* [*Free Regsitration Required]|
This rear-panel connector is used in the following modes: This error is also reported when a character-type parameter is too long.
From the remote interface, you can also display a character message on the front panel. Unsecure or secure the instrument for calibration. The default is 1 ms. The function generator beeps once each time an error is generated unless you have disabled the beeper.
You can select one of the five built-in arbitrary waveforms, one of four agjlent waveforms, or the waveform currently downloaded to volatile memory.
The default is POS rising edge. Select whether the function generator uses the rising edge or falling edge of the trigger signal on the rear-panel Trig In connector for an externally-triggered sweep. Page Chapter 3 Features and Functions Agilnet Overview Calibration Message The function generator allows you to store one message in calibration memory in the mainframe.
Simplified Programming Overview Chapter 4 Remote Interface Reference Simplified Programming Overview Simplified Programming Overview This section gives an overview of the basic techniques used to program the function generator over the remote interface.
Features And Functions Features and Functions This chapter makes it easy to look up all the details about a particular feature of the function generator. Check the list of parameters to verify that you have used a valid parameter type.
Chapter 3 Features and Functions Frequency Sweep Sweep Trigger Source In the sweep mode, the function generator outputs a single sweep when a trigger signal is received. Select whether the function generator uses true-high or true-low logic levels on the rear-panel Trig In connector for an externally-gated burst. If you set the carrier to a value less than the deviation frequency with FM enabledthe function generator will automatically adjust the deviation to the maximum value allowed with the present carrier frequency.
Chapter 7 Tutorial Pulse Waveform Generation Pulse Waveform Generation To eliminate distortion due to aglient at higher frequencies, the Agilent A also uses a different waveform generation technique to create pulse waveforms.
Agilent Technologies 33220A User Manual
Page 50 characters. The Sync signal always goes from low to high at the beginning of the sweep. Chapter 3 Features and Functions Pulse Width Modulation PWM Width Deviation The width deviation represents the variation in width in seconds agilentt the modulated waveform from the width of the original pulse waveform.
The variation in phase of the modulated waveform from the carrier waveform is called manuaal phase deviation, which can vary from 0 to degrees. You can kanual the arbitrary waveform directly from volatile memory or you can copy the waveform to non-volatile memory using the DATA: 332220a characters are converted to upper case. You may want to disable the front-panel beeper for certain applications. The CD auto-starts and provides a Welcome page with instructions. The following steps show you how to change the frequency to 1.
The function generator takes the specified number of points and expands them to fill waveform memory. Used only when the Internal modulation source is selected FM: This connector accepts TTL-compatible levels and is referenced to chassis ground not floating ground. Used in the internal triggered burst mode only. Burst Chapter 7 Tutorial Burst connector can be configured as an output to enable the A to trigger other instruments at the same time as its internal trigger occurs.
AGILENT TECHNOLOGIES A USER MANUAL Pdf Download.
The commands used for sweep are listed on page This section describes the low-level commands used to program the function generator to output a pulse waveform. The isolation between chassis ground circuits and floating circuits is controlled by an optical isolation barrier and a serial link. Chapter 5 Error Messages Execution Errors Settings conflict; high level changed due to low level You can set the agilebt to a positive or negative value, but note that the high level must always be qgilent than the low level.
Press the Burst Period softkey and then set the period to 20 ms using the numeric keypad or the knob and cursor keys.
The APPLy command provides the most straightforward method to program the function generator over the remote interface. Chapter 3 Features and Functions Calibration Overview To Unsecure for Calibration You can unsecure the function generator either from the front panel or over the remote interface.
When you enable a modulation, sweep, or burst mode, all other modes are turned off. We have written the exercises in this chapter to prepare the instrument for use and help you get familiar with some of its front-panel operations. Spectrum of Above Waveform at kHz Square Waveform Generation To eliminate distortion due to aliasing at higher frequencies, the A uses a different waveform generation technique to create square waves.
Output Configuration Chapter 3 Features and Functions Output Configuration Output Configuration This section contains information to help you configure the function generator for outputting waveforms. The remainder of this chapter contains examples of using the front-panel menus. Pulse Width Modulation PWM PWM is used in digital audio applications, motor control circuitry, switching power supplies, and other control applications. Chapter 5 Error Messages Execution Errors Settings conflict; edge time decreased due to pulse width For a pulse waveform, the function generator will automatically adjust the waveform parameters in the following order as needed to generate a valid pulse: Instructions and mounting hardware are included with each rack-mounting kit.
To configure the function generator to recall mabual factory default state at power-on, press and then select the Pwr-On Default softkey. Used only when the Internal modulation source is selected PWM: Chapter 5 Error Messages Execution Errors Settings conflict; selected arb is missing, changing selection qgilent default If you delete an arbitrary waveform from non-volatile memory after storing the instrument state, the waveform data is lost and the function generator will not output the waveform when the state is recalled.
To avoid multiple waveform changes, you can enable the burst mode after you have set up the other burst parameters. With autoranging disabled, the function generator uses the current amplifier and attenuator settings.
The default is OFF. The following steps show you how to change the offset to —1. The locations are numbered 0 through 4.